Webmaster’s blog 2019
Alan Wood’s microscopical diary
My parents often used to take my sister and me to Wimbledon Common on the 93 bus from North Cheam, and I just about remember carrying my jam jar and fishing net and coming home with small fish or tadpoles from Rushmere. That was 60 years ago, and it is great to be going there again with the Quekett for the excursion, BioBlitz and Open Day. For the BioBlitz on 22nd and 23rd June this year, the weather had only just started to get warm and there were nowhere near as many butterflies and damselflies as last year. I used my Canon EOS 40D with the Canon 60 mm f/2.8 macro lens and Yongnuo YN-14EX Macro Ring Lite and got a reasonable success rate for images to go on the website.
Painted lady butterfly
I am reminded of being in my late teens and early twenties with a Miranda GT, 40 mm f/2.8 Makro-Kilar lens and a Sunpak ringflash powered by a pack of batteries slung from my shoulder. Now I would not like to be without auto-focus, automatic stop-down and automatic TTL exposure.
A day out on Box Hill on 26th June was completely different, swarming with marbled whites and meadow browns, and we saw several painted ladies and silver-washed fritillaries, but far fewer orchids than last year. We also saw a hummingbird hawk-moth, a first for me, an amazing sight with its wings just a blur.
We moved nearly 2 months ago, but it has taken us a long time to get settled. I have only just started using my microscopes again, because I wanted a photomicrograph to use in promoting the “Arranged specimen slides” gossip on Facebook. The only arranged slide I have is by Brian Darnton, one of the ones that I bought from him at Langton Matravers in April.
Stellate Art Work by Brian Darnton
The arrangement is about 3 mm wide, so to get it all in I used a Carl Zeiss Plan 2.5/0,08 objective and an Olympus NFK 2.5× LD photo eyepiece. The plant hairs and fern scales are colourless, and I used crossed polarisers and a retarder cut from the window on a cake box to produce the colours in the photomicrograph.
Moving has provided an opportunity to organise my 5 microscopes and lots of accessories accumulated over the last 15 years, and now almost everything is in a couple of cupboards right next to my desk.
Trinocular Olympus BHT with Canon EOS 5D Mark II controlled from EOS Utility on my desktop computer
In the interests of domestic harmony, I have to put my microscopes away when I am not using them. In the flat, I had to lift my BH-2 from floor level and carry it to another room to use it; in the bungalow it is stored at the right height and I only have to move it from the cupboard top the adjacent desk.
At the PMS Husbands Bosworth meeting on 11th May there was a USB cable for a Nikon camera on Mike Smith’s “free to a good home” table so I thought I would try it with my uncooperative Canon cameras. I tried it when I got home, and suddenly my EOS 40D and 600D were recognised by my computer again. Looking closely, it appeared that the plug on the Nikon cable went further into the socket on the side of the cameras than the Canon plug. It turned out that I was not using enough force to push in the Canon cable, and so it did not connect properly. A bit more force and the Canon cable went all the way in and worked!
Very little time for microscopy because we are busy moving from a flat to a bungalow, but I still get to as many meetings as I can. A new member who lives near Langton Matravers asked for help with photomicrography and with using crossed polarisers, so I took my trinocular Olympus CH-2, a laptop with EOS Utility, my EOS 40D body and some filters to the meeting on 13th April. I have used this equipment lots of times, but on this occasion the computer would not recognise the camera. I tried the 40D and the 600D and 2 different USB cables, but with no success. Back home I tried them again, on my laptop and on my desktop computer, and they still wouldn’t connect. My EOS 5D Mark II worked fine. I spent hours on the Internet trying to find a solution, but nothing worked.
Kai with our trinocular Olympus CHS
Brian Darnton was selling slides that he has made, some of forams and some of arranged plant hairs and fern scales, so I bought a couple, but I don’t know when I will get a chance to look at them.
At the workshop on diatoms from Lake Mývatn I followed my normal procedure of taking lots of photographs so that I can pick out the best ones and write the meeting report around them. However, this time the card failed while I was copying the images to my hard disk, and I only managed to recover 3 photos. I don’t try to economise by buying unknown brands, this was a genuine SanDisk Extreme PLUS SD card that I have been using since 2016. SanDisk are going to replace it under guarantee. This card was already a replacement under guarantee, after my previous SanDisk Extreme failed after only a year. Now I understand why some cameras aimed at professionals have 2 card slots, so you have a backup.
My first microscope was a Watson Kima, bought second-hand when I was still at grammar school. I have kept it for over 50 years, but now I need the space so I reluctantly decided to give it away. I have used it occasionally over the years and it has always worked fine, but this time the fine focus didn’t work properly – it either did nothing or else moved much too far. I asked for help in the Amateur Microscopy group on Facebook and in the Microscope group on Yahoo! and Merv Hobden and SaraLee Shepherd both replied promptly with the same diagnosis and treatment, and they were correct. The fine focus knobs move a block up and down in a dovetail slider, and the grease had hardened.
With the body tube removed, it should be possible to lift the block a little and it should slide down again. I could lift it, but it didn’t slide down. Following advice, I undid two screws holding the plate on top of the block, being careful not to lose the spring below it. Then I slide the block up and out of the dovetails. I used Ronsonol lighter fluid to clean off the old grease and then applied a thin smear of Vaseline. Everything went back together easily, and the fine focus know works again.
I offered the working microscope in the Amateur Microscopy group and the Microscope Wants, Sales And Services group on Facebook, free to anyone who could collect it, and it now has a new home.
Watson Kima No. 95030
Before parting with it, I added the serial number to the Watson Microscope Stand Dating Table on the Quekett website.
About this blog
I am Alan Wood, webmaster for the Quekett website, and author of several web pages on Olympus microscopes. I spend too much time writing about microscopes and buying more equipment. I hope this blog will help me to focus on using my microscopes so that I have something to write about!
If you would like to comment on anything in this blog, please contact the Webmaster.